Programmable discharge voltage 200 V to 16.5 kV Stabilized charging...
Ideal for voltage bleeding to Ground Plane during Indirect ESD Immunity...
Rentals Ship Immediately Purchase Options See: Used Keytek Minizap ESD...
Suitable for EN/IEC 61000-4-2 ESD immunity compliance testing...
ESD Guns can refurbish your existing INA 405 battery pack. Expected...
Rental Package Includes Air and Contact Discharge Tips Output Capability...
Air- and contact-discharge to 30 kV Color touch panel control Advanced...
Flexibility – enables users to meet many different test specifications...
Notice: Mil-STD-461G CS118 is a brand new core requirement for Mil-STD-461 testing. Mil-STD-461G is in a Draft State as of March 2015, adding the electrostatic discharge testing section of conducted susceptibility hereon referred to as CS118. These revisions aren't expected to take affect until the end of the year. In general, the ESD levels, test setup and test equipment are equivalent to that of IEC 61000-4-2 and RTCA/DO-160.
This requirement is applicable to electrical, electronic, and electromechanical subsystems and equipment that have a man-machine interface. It is not applicable to equipment that interfaces with or controls ordnance items.
The EUT shall not exhibit any malfunction, degradation of performance, or deviation from
specified indications, beyond the tolerances indicated in the individual equipment or subsystem
specification, when subjected to the values shown Table CS118-1. Contact up to 8kV is the
required method. Air discharges shall be required only if specified by the procuring activity.
5.16.3 CS118 test procedures
This test procedure is used to verify the ability of the EUT to withstand personnel borne
electrostatic discharge (ESD) in a powered up configuration.
A ground reference plane shall be provided on the floor of the laboratory. The EUT shall be connected to the grounding system and arranged and connected according to its installation specific cations. A distance of 1 m minimum shall be provided between the EUT and any metallic structure. The discharge return cable of the test generator shall be connected to the ground reference plane, and this connection shall be of low impedance.
In cases where the length of the cable exceeds the length necessary to apply the discharges to the selected points, the excess length shall be placed non-inductively off the ground reference plane and shall not come closer than 0.2 m to other conductive parts in the test set-up.
Full Version (Released Dec. 11,2015): MIL-STD-461G
Data shall be presented detailing each discharge point, tip used, voltage applied to the EUT at that point (including magnitude and polarity), type of discharge (air or contact), and resulting operating behavior.
Ground reference plane: A copper or aluminum sheet of 0.25 mm minimum thickness: other materials may be used but they shall have at least 0.65 mm minimum thickness. The minimum size is 1 m2. The exact size depends on the EUT. It shall project beyond the EUT or coupling plane by at least 0.5 m on all sides. It shall be connected to the protective earth.
Coupling planes: These planes shall be constructed from the same material and thickness as that of the ground reference plane and shall be connected to the ground reference plane via a cable with a 470kΩ resistor located at each end.